Payment & Shipping Terms:
|IEC60884, IEC61058, IEC60669, IEC60320
|Linear: 0～150mm, Can Be Preset
|Linear: 0～200mm/s, Can Be Preset
|0～999999 Times, Can Be Preset
|0～999.9s, Can Be Preset
|6 Station Linear
IEC60884-1 Six Stations Switches Plug and Sockets Endurance Tester
Standards & Clauses:
This apparatus conforms to IEC60884-1:2006 “Plugs and socket-outlets for household and similar purposes- Part1: General requirements” clause 20 and 21
IEC61058-1:2008 “Switches for appliances Part1: General requirements” clause 17
IEC60669-1:2007 “Switches for household and similar fixed-electrical installations - Part1: General requirements” clause 18 and 19
IEC60320-1:2007 “Appliance couplers for household and similar general purposes- Part1: General requirements” clause 19 and 20.
|6 station linear
|7-inch color LCD touch screen
|0～999999 times, can be preset
|0～999.9s, can be preset
|0～999.9s, can be preset
|Linear: 0～150mm, can be preset
|Linear: 0～200mm/s, can be preset
|Atmospheric pressure 80～106kpa; Ambient temperature 5～40°C; Relative humidity (20～90)%RH; Use place: no violent shaking, vibration, electromagnetic interference, dust, explosive and corrosive gas, good heat dissipation
Every Other Output Function
This tester has function: Every other output function, which is mainly used in the plug and socket test. After the load current is connected and the current detection function is turned on, when the plug is inserted into the socket for the first time, the program controls the current of the load cabinet to flow through the socket, and then closes the load cabinet after pulling it out. When the plug is inserted into the socket for the second time, no current flows through the socket, there is current for the third time, and there is no current for the fourth time. In this cycle, the purpose of intermittent power-on is achieved.
Please read the product instruction manual carefully before using the equipment, master the knowledge of electricity safety, and take corresponding safety precautions; this test can be carried out only when it is confirmed that there is no potential safety hazard.
The voltage passing through the sample is 1.1 times the rated voltage of the sample, and the current passing through the sample is 1.25 times the rated current value of the sample. Please select the voltage and current value according to the standard, otherwise it will cause damage to the sample and accidents that endanger personal safety.
Please ensure that the wire connecting the sample switch to the load is compatible with the allowable rated current of the sample switch.
During the test work process, please observe the operator carefully and do not leave the operation post.
The wiring of the terminal on the rear panel of the instrument should be in reliable contact and ensure good insulation.
After power on, do not touch the terminal on the rear panel of the instrument to prevent electric shock.
When replacing the fuse, remember to cut off the power supply before you can replace it with a fuse of the same capacity.
Contact Person: Ms. Zoe Zou
Tel: +86 13527656435