Payment & Shipping Terms:
|Applied Standard:||YDT757||Test Sample:||Microwave Tower|
|Handle Material:||Oak||Hammer Weight:||70g|
|Hammer Mass:||210g±1g||Hammer Material:||45 Steel|
|Hammer Hardness:||Above HS40||Base Size:||250×250×15mm|
Coating Adhesion Hammer Test Device,
Oak Handle Hammer Test Device,
Oak Handle Hammer Test Device
YDT757 Galvanized Coating Adhesion Hammer Drop Test Device
This test device conforms to the YDT757 drop hammer test method, and is suitable for hot-dip galvanized coating adhesion test. It uses a hammer to drop freely to hit the sample to test whether the galvanized coating meets the standard. This product is suitable for product quality inspection and control of similar iron tower manufacturers and quality inspection departments.
Sample and Restriction:
Transmission line tower, microwave tower.
It is used to test hot dip galvanized coating adhesion.
Drop freely impact samples of the hammer head to test galvanized Layer meets standards or not.
Adjust the device according to the countertop level to make it level, Adjust the sample so that the striking point is not less than 10mm away from the edge, corner, and end of the sample, the hammer head faces the center of the platform, the hammer handle is perpendicular to the base plane, and then falls freely, striking 5 points in parallel at 4mm intervals. Check whether the surface condition of the galvanized layer meets the requirements of the standard.
|Hammer hardness||Hardness above HS40|
|Vertical height from center line of hammer head to base surface||300mm±1mm|
|Hammer handle material||Oak|
|Hammer handle mass||70g|
|Base plate||250 ×250×15mm, equipped with horizontal level bubble|
Contact Person: Ms. Zoe Zou
Tel: +86 13527656435