Payment & Shipping Terms:
|Applied Standard:||IEC 60227.2, IEC 60245.2, Etc.||Applied UUT:||Rubber Plastic Wires And Cables|
|Test Station:||Single Station||Load Current:||0.5~25A Adjustable|
|Testing Distance:||1000±20mm||Load Voltage:||L-L 10~400V Adjustable|
|Testing Speed:||0.33m/S||Suitable Cores:||2-18|
IEC 60227.2 Flexing Test Apparatus,
0.33m/S Flexing Test Apparatus
Rubber Cables Flexing Test Apparatus IEC 60227.2 Figure 1 0.33m/S
The test machine is designed and manufactured according to standard IEC 60227.2 Clause 3.1 Figure 1, IEC 60245.2 Clause 3.1 Figure 1, IEC60245-1 clause 220.127.116.11 and etc. It complies with the relevant standard requirements of rubber plastic wires’ and cables’ mechanical strength flexing test. This test does not apply to flexible cables with cores of nominal cross-sectional area greater than 4mm2 and cables having more than 18 cores laid up in more than two concentric layers.
The test apparatus can meet the standard test requirements of the domestic and international standards, can be widely used. It has precision digital meter, thus, can accurately measure the smaller or larger current voltage. The change method for pulleys is simple and convenient. The test equipment adopts to chain drive mechanism, the performance is stable and the noise is rare.
|Power Supply||3 phase 5 wires 220VAC/ 60 Hz or 380V/50Hz|
|Load Current||0.5~25A, adjustable|
|Load Voltage||Line-to-line voltage 10~400V adjustable|
|Nominal cross-sectional area||0.2~4mm2|
|Diameter of pulleys||60, 80, 120, 160, 200mm|
|Test weights||2*0.5kg with hook, 2*0.5kg with hole, 6*1kg with hole, 2*6kg with hook|
How to installand adjustof the pulleys?
Contact Person: Manager