Payment & Shipping Terms:
|Applied Standard:||IEC 60950||Applied Clause:||Clause 2.3.5|
|Power Fuse:||3A||Output Frequency:||50Hz/60Hz±2words|
|Internal Impedance:||1200Ω±2%||Output Power:||25W|
|Output Voltage:||120V±2V||Input Frequency:||50Hz/60Hz|
Clause 2.3.5 Information Technology Equipment,
IEC 60950 Information Technology Equipment,
130A Test Generator
IEC 60950 Clause 2.3.5 Information Technology Equipment 5 Safety Test Generator 130A
The 130A test generator is designed and manufactured according to the requirements of IEC60950 clause 2.3.5.
It can simulate voltage standards (50Hz or 60Hz) in different regions to meet the test requirements. The instrument is designed with a stable signal source and a large and medium power matching circuit to meet the output requirements under large loads.
|Input voltage||AC 120V±10%|
|Working environmental||Temperature:-20℃～40℃, Humidity: < 80％ RH|
|Dimensions||Approx. 408mm (L) * 320mm (D) * 180mm (H)|
|Function||With commutation connection requirements|
The 130A test generator is packed in a carton box. When unpacking the instrument, check the appearance and panel are intact, and check whether the number of instrument accessories in the box is correct. For the list of instrument accessories, see packing list.
Input voltage and fuse specifications
The 130A test generator uses AC220V, 50Hz/60Hz single-phase power supply, and the fuse capacity is 3A. For details, see the technical specifications. Note: Before replacing the fuse, the input power cord must be removed, and the newly replaced fuse must meet the requirements.
Before turning on the power of the instrument, make sure that the ground wire of the power supply is connected. This instrument uses a three-core power cord. When the power cord is connected, it must be in good contact, and ensure that the instrument chassis is grounded.
Storage and transportation environment
The instrument can be stored and transported in the following environmental conditions:
Humidity: less than 80% RH.
Note: It is necessary to avoid abrupt changes in ambient temperature, which may cause moisture condensation inside the instrument.
Instructions of Panel :
①: Power switch (POWER)
②: Output frequency selection
③: Commutation selection
④: Output voltage display
⑤: Output voltage terminals
Contact Person: Manager