Payment & Shipping Terms:
|Test Sample:||Double/single Handle Cookwares||Applied Standard:||BS EN 12983-1|
|Applied Clause:||Clause 7.5 And Appendix D||Standard Figure:||Figure D.1|
|Test Contents:||Bending Strength Test||Appled Figure:||Figure D.1|
|Loading Bar Diameter:||Ф10±0.1mm||Range Of Force:||1-500N|
EN 12983-1 Bending Resistance Test Equipment,
500N Bending Resistance Test Equipment,
Clause 7.5 Bending Resistance Test Equipment
BS EN 12983-1 Double Handle Cookware Bending Resistance Test Equipment
The design of the cookware handle bending strength test apparatus of cookware complies with BS EN 12983-1:2020 clause 7.5 and Appendix D, figure D.1.
It is used for the bending strength test of the handle of the cookwares. The fastening connecting part of the handle bears a force of 100N, and there should be no failure in the joints such as riveting and welding.
|Loading bar type||D-section|
|Loading bar diameter||Ф10±0.1mm|
|Range of force||1-500N|
|Dimensions||Length * width * height: 1000mm*500mm*810m|
|Power supply||Match client's local power supply requirements|
Install the cookware as shown below, adjust the left and right, up and down positions of the push-pull force gauges to align the D-section loading bar(s) with the measured handle(s) position. If testing single-handle cookware, make sure that the distance between the contact surface of the D-section loading bar and the handle is 10±1mm from the end of the handle. If testing double-handle cookware, the positions of push-pull force gauges on both sides should be well adjusted.
Contact Person: Manager