Payment & Shipping Terms:
|Standard Clause:||IEC60884 Clause 24.9||Test Figure:||Figure 29|
|Drop Height:||750mm||Test Times:||8 Times|
|Rotation Action:||Every 45°||Operation Method:||Manual Release|
switch endurance tester,
switch life testing machine
IEC 60884-1 Apparatus for Mechanical Strength Test on Multiple Portable Socket-outlets
The socket-outlet mechanical strength test apparatus meets the requirements of IEC 60884-1 2013 clause 24.9 and figure 29. The apparatus is used to test the mechanical strength of multiple portable socket-outlets when rewirable multiple portable socket-outlets are fitted with the lightest type of flexible cable of the smallest nominal cross-sectional area.
|Distance to fixed end||2250mm|
|Test Times||8 times|
|Rotation action||Every 45°, 8 directions can be rotated|
|Operation method||Manual release|
|Conforms to standard||IEC 60884-1 2013 clause 24.9 and figure 29|
The free end of the flexible cable is fixed to a wall at a height of 750mm above the floor, as shown in figure 29.
The specimen is held so that the flexible cable is horizontal and then it is allowed to fall onto a concrete floor, eight times, the flexible cable being rotated through 45° at its fixing after each fall.
After the test, the specimen shall show no damage within the meaning of this standard; in particular, no part shall have become detached or loosened.
The fixtures for the specimen on the two ends of this apparatus
Contact Person: Manager