Payment & Shipping Terms:
|Push Force Gauge:
|Pull Force Gauge:
|Height Gauge Pre-measurement
|Aluminum Profile Production
Semi Luminaire Bending Moment Test Equipment,
IEC 60598-1 Bending Moment Test Equipment,
Lampholders Bending Moment Test Equipment
IEC 60598-1 Semi-Luminaire Bending Moment Test Equipment For Screw Or Lampholders
Product Information For Semi-Luminaire Bending Moment Test Equipment:
It is conforming to IEC 60598-1clause 4.14.2, Mainly used to detect the bending moment of a semi-luminaire intended to be connected to a screw or bayonet socket due to its own weight.to Test the bending moment of the lamp in the lamp holder. The device adopts aluminum profile frame, imported push-pull force meter and precision linear guide rail. It has beautiful appearance, stable movement and small clearance to ensure accurate test.
Technical Parameters For Semi-Luminaire Bending Moment Test Equipment:
1. Push and Pull Force Gauge 0-10N.m,Minimum scale 0.1N.m, accuracy±0.3%F.S； IMADA Push and Pull Force Gauge FB-10N(optional )
2. Lifting mechanism:0-100mm,Linear guide, manual turntable；
3. The bending moment and the weight of the lamp itself are shown in Table 2.
4. Force arm: Height gauge pre-measurement
5. Frame: Aluminum profile production
Test Principle For Semi-Luminaire Bending Moment Test Equipment:
As structure figure below : Supporting the lamp head through the thin metal piece on the left ,use pull and push gauge hang the top of the lamps, Shake the lift handle, make the push-pull gauge gradually rise to the level of the lamp, record the pull and push gauge’s display data , This force value is multiplied by the force arm (the distance from the support point on the left side of the lamp to the connection point on the right side of the lamp) is the bending moment applied by the lamp to the lamp holder. This bending moment shall be less than the value specified in Table 2. Note: The weight of the lamp itself should not exceed the weight given in Table 2.
Contact Person: Ms. Zoe Zou
Tel: +86 13527656435