Payment & Shipping Terms:
|Standard Clause:||IEC60884 Clause 14.23.2||Test Sample:||Plug|
|Balancing Arm:||0-250mm||Hang Weight:||2.5N|
Plug Torque Force Test Device,
IEC 60884-1 Force Test Device,
Clause 14 Force Test Device
IEC 60884-1 Clause 14 Plug Torque Force Test Device 0.25 & 0.4 N.m
This device can measure the torque of the plug inserted into the socket according to the standard.
The purpose is to prevent the device that is intended to be inserted into a fixed output socket and equipped with a pin to not subject the output socket to excessive stress.
Mainly detect fixed output sockets and devices equipped with pins
Standads & Clauses:
IEC60065:2005clause 15.4.1 and figure 11
Insert the device into a fixed socket conforming to this standard. The socket is rotated 8 mm behind the socket joint face by a horizontal axis through the axis of the live socket contact and parallel to the joint face.
The additional torque that must be applied to the socket to maintain the joint in the vertical plane shall not exceed 0.25 Nm.
|Specimen Eccentricity||Distance from the contact surface and the Axis of the surface:8mm|
|Weight||2.5N, can achieve 0.25Nm to 0.4Nm torque force|
|Torque Calculation Method||G×L(such as 0.25N.m,use 2.5N weights and 100 mm arm)|
|Configure||One of each universal plug (suit for GB and BS plug), UL plug, European plug|
Contact Person: Manager