Payment & Shipping Terms:
|Control Method:||PLC Control||Drive Mode:||Electrical|
|Flexing Angle:||0-360° Can Be Set||Flexing Speed:||0-60cycles/min Can Be Set|
|Loading Weight:||5N,10N||Loading Voltage:||0-250V Can Be Regulated|
|Applied Standard:||IEC 60335-1, IEC 60884-1, Etc.||Product Name:||Flexible Cable Flexing Testing Equipment|
IEC 60335-1 Electrical Appliance Testing Equipment,
Electrical Flexible Cable Testing Equipment
IEC 60335-1 Programmable Electrical Appliance Flexible Cable Flexing Testing Equipment
Standards & Clauses:
This flexible cable flexing testing equipment meets the requirements of IEC 60335-1 clause 23.3 and 25.14, IEC60335-2-14, IEC60884-1 clause 23.4, IEC 60745-1 clause 24.12 and etc.
It is used to check that the supply cord of the small household appliances (such as Coffee pot / electric iron) and lamps is adequately protected against excessive flexing where it enters the appliance. It is suitable for testing the flexible Small household appliances, such as coffee pot, electric iron, toaster, luminaires and etc.
The test device adopts to PLC electrical control, 7 inch color touch screen intelligent operation system; servo motor controlling the test angle, high precision digital instrument, high performance digital frequency conversion, precise speed regulation, six digits memory timer, multifunctional combined clamp, center gage of rotation, bending angle adjustable, 4kw output power. Test times (maximum of 999999 times), test speed, and test angles can be preset; it has electric current detection function. This device can meet the test requirements of most of small household electrical appliances, the applicable range is widely.
|Input Power||Match client's local power supply requirements|
|Control Method||PLC control, touch screen operation|
|Station||One (can be customized)|
|Flexing Angle||0-360° (±30°, ±45°, ±60°, ±90° can be preset on touch screen )|
|Flexing Speed||0-60cycles/min, can be preset|
|Test Cycles||0-999999, can be preset|
|Loading Weight||5N,10N (can be customized)|
|Loading Voltage||0-250V, can be regulated|
|Sample judgment||Judge the conduction by current|
How to fix the test sample on the fixture?
Select appropriate length screw rods according to the size of test sample. Planning the position of installation position on rotating disc.
The installation principle of the sample is that the power cord of the sample and the power cord connection position in the center axis of rotating disc.
When the supply cord is at the starting position, the axis of the cord where it enters the cord guard or inlet is vertical and passes through the axis of manipulator, at this time, then user can determine the clamp bolt mounting hole position, two mounting screw rods with nuts are installed on the rotary disc, then fastening nuts, nuts can be added according to the shape of the specimen. Install the base plates of the fixture on the screw rods, and adjust their positions along the screw rods, after that, install the sample leaning on the base plates of the fixture, then install the other two plates of the fixture to clamp the sample well, make sure that the fixture and test sample are installed well and will not affect flexing.
Contact Person: Manager