Payment & Shipping Terms:
|Standard:||IEC60669||Clause:||12.3.12 And Figure 11|
|Test Station:||Single||Deflection Angle:||0-360°|
|Bending Force:||0.25 N, 0.5 N, 1.0N||Operation Method:||Manual|
IEC60669 Clause 12.3.12 and Figure 11 Voltage Drop Tester for Plugs and Socket-Outlets
The Voltage Drop Tester is designed and manufactured according to the standard requirements of IEC60669-1: clause 12.3.12 and figure 11, IEC60884-1: clause 12.3.12 and figure 12 and etc.
It is used to check screwless terminals shall be so designed that the connected rigid solid conductor remains clamped, even when it has been deflected during normal installation, for example, during mounting in a box, and the deflecting stress is transferred to the clamping unit.
|Clause||12.3.12 and figure 11|
|Bending Direction||Towards any one of the 30°±3° difference, a total of 12 directions|
|Bending Force||0.25 N, 0.5 N, 1.0N|
|Direction of Application||Perpendicular to the direction of the unbending wire|
|Judgement Criterion||Voltage-drop not greater than 25 mV or greater than 1 min|
Contact Person: Manager