Payment & Shipping Terms:
|Equipment Dimensions:||1600 × 1200 × 2000||Equipment Weight:||180kg|
|Power Supply:||AC220V50Hz||Working Temp.:||5-35℃|
|Force Speed:||0-200mm/s||Test Range:||Single And Multi-position|
Carrying Handle Strength Test Device,
IEC62368-1 Carrying Handle Strength Test Device
IEC62368-1 IEC62368-1 LCD Screen Operation Single Station Carrying Handle Strength Test Device
Carrying Handle Strength Test Device conform to IEC62368-1:2018 clause 8.8,IEC60601-1:2012
Mechanical strength of handles for audiovisual, information technology and communication technology equipment and portable medical electrical equipment with carrying handles or handles.
|Equipment dimensions||about: 1600 × 1200 × 2000 (mm) W × D × H|
|Constant force time||0-9999s,Can be set|
1.Equipment structure and characteristics: Equipment dimensions: about: 1600 × 1200 × 2000 (mm) W × D × H, equipment weight: 180kg
2.Use environment: Power supply: AC220V50Hz, working temperature 5-35℃, humidity 30%-90%;
3.Control and operation: LCD screen operation (optional PLC control, touch screen operation);
4.Station: single station, frame: Aluminum profile rack, suitable for fixing different samples, Protect: Overload protection, leakage protection;
5.Dimensions and weight: WDH about 1600X1200X2000mm, about 180kg.
Specimen and Limitations:
Audio-visual, information technology and communication technology equipment and portable medical electrical equipment with carrying handles or handles.
Apply a force equal to 1~4 times the weight of the equipment to the handle through the PLC control force device, and evenly apply the force to the center of the handle in a range of 75mm (7cm for ME equipment), do not yank, start from zero within 5s-10s Gradually increase to the test value, keep for 1min.
GB4943.1-2022Section 8.8 Requirements；
IEC62368-1:2018 clause 8.8；
Contact Person: Manager