Payment & Shipping Terms:
|Applied Standard:||IEC60065:2014||Applied Clause:||Clause 22.214.171.124, 6.2.1|
|Probe End Radius:||R6 Mm||Probe Diameter:||Φ12 Mm|
|Baffle Diameter:||Φ50 Mm||Probe Length:||80 Mm|
|Probe Material:||Stainless Steel||Handle Material:||Nylon|
Stainless Steel Blunt Test Probe,
IEC 62368-1 Blunt Test Probe,
Figure V.3 Blunt Test Probe
IEC 62368-1 R6mm Stainless Steel Blunt Test Probe Of Figure V.3
The blunt probe of figure V.3 conforms to IEC 62368-1:2018 clause 5.3.2, 5.4.10, V.1.4 and figure V.3, IEC60950 figure 2C, IEC60065 figure B1, IEC62151 figure 3 and UL6500 figure B.1.
It is used for the bare parts of TNV circuits of information technology equipment to see whether they have sufficient protection.
|Probe Length||80 mm|
|Probe End Radius||R6 mm|
|Probe Diameter||Φ12 mm|
|Baffle Diameter of Handle||Φ 50 mm|
|Probe material||Stainless steel|
Sinuohave a full range of test probes according to different standards, and also can offer customized service.
Test procedures and operation
Please refer to the relevant standards.
Take IEC 60950 clause 126.96.36.199 as an example:
This test probe is applied to openings, without appreciable force, in every possible position, except that floor-standing equipment having a mass exceeding 40 kg is not tilted.
The test probe doesn’t have built-in force, if specific applied force required, the test probe should work together with a force gauge. At the end of our probe handle, there is a M6 thread hole to match the force gauge. Showing as following reference picture:
1. Please do not damage the appliance shape (such as scratching, bending, etc.).
2. Prevented from dust and moisture environment, in order to avoid the oxidation to affect the dimensional accuracy.
3. Be careful of electric shock.
Contact Person: Ms. Zoe Zou
Tel: +86 13527656435