Payment & Shipping Terms:
|Applied Standard:||IEC 62368-1||Clause:||4.7|
|Test Station:||Single||Balancing Arm:||L:0-250mm|
|Weight:||2.5N, Can Achieve 0.25Nm To 0.4Nm Torque Force||Specimen Eccentricity:||Distance From The Contact Surface And The Axis Of The Surface:8mm|
Plug Torque Force Test Apparatus,
IEC 62368-1 Force Test Apparatus,
Clause 4.7 Force Test Apparatus
IEC 62368-1 Clause 4.7 Plug Torque Force Test Apparatus 0.25 & 0.4 N.M
This device can measure the torque of the plug inserted into the socket according to the standard of IEC 62368-1:2018 clause 4.7, IEC60065: 2014 clause 15.4.1 and figure 11, IEC 60884-1 2013 clause 14.23.2, IEC 60335-1 clause 22.3, IEC60065 clause 15.4.1 and figure 11 and etc.
The purpose is to prevent the device that is intended to be inserted into a fixed output socket and equipped with a pin to not subject the output socket to excessive stress.
Mainly detect fixed output sockets and devices equipped with pins
Insert the device into a fixed socket conforming to this standard. The socket is rotated 8 mm behind the socket joint face by a horizontal axis through the axis of the live socket contact and parallel to the joint face.
The additional torque that must be applied to the socket to maintain the joint in the vertical plane shall not exceed 0.25 Nm.
|Applied Standard||IEC 62368-1|
|Specimen Eccentricity||Distance from the contact surface and the Axis of the surface:8mm|
|Weight||2.5N, can achieve 0.25Nm to 0.4Nm torque force|
|Torque Calculation Method||G×L (such as 0.25N.m, use 2.5N weights and 100 mm arm)|
|Configure||One of each universal plug (suit for GB and BS plug), UL plug, European plug|
Contact Person: Manager