Payment & Shipping Terms:
|Test Station:||Single Station||Electrical Control:||PLC|
|Drive Mode:||Servo Motor||Rotation Angle:||0-360° Adjustable|
luminaire testing equipment,
led light testing equipment
IEC 60598-1 Luminaries Adjusting Devices 360° Movement PLC Control Test Apparatus
Standards & Clauses:
IEC 60598-1:2014 “Luminaries-Part1: General requirements and tests” clause 4.14.3
Samples and Restrictions:
Adjusting devices or means of adjustment, for example joints, hoisting devices, adjusting brackets or telescopic tubes.
To verify adjusting devices or means of adjustment, for example joints, hoisting devices, adjusting brackets or telescopic tubes, shall be so constructed that cords or cables are not pressed, clamped, damaged or twisted along the longitudinal axis by more than 360° during operation.
Adopts the motor to drive the turntable, then simulates normal operation of lamp adjusting devices and then reset.
The adjusting devices, equipped with the appropriate cable or cord shall be operated in according with IEC 60598-1 : 2014 table 4.5. A cycle of operation is a movement from one extreme of the range to the other and back to the starting position. The rate of movement shall not cause the device to hear appreciably and shall not exceed 600 cycles per hour.
Divided into electrical control part and fixture part
|Test station||Single station|
|Electrical control method||PLC intelligent integrated control, 7-inch human-machine touch interface operation|
||Servo motor drive|
|Sample clamping method||
Mechanical clamping, the clamping point position is adjustable
|Rotation angle||0-360° arbitrarily adjustable, can be rotated on one side or on both sides|
|Maximum radius of gyration of the pendulum||R500mm|
|Test speed||0-10 times / min (<600 times / h)|
|Test times||0~999999 presettable, a cycle of operation is a movement from one extreme of the range to the other and back to the starting position.|
Contact Person: Manager